I have been waiting for a long time for a usable low-cost spectrum analyzer and it looks like that wait is over.
I bought a TinySA spectrum analyzer for $55 from R&L Electronics (one of the official dealers). If you get one, try to use the official sites; there are lots of bad clones out there. The TinySA is, as the name implies, a tiny, battery-powered, touch-screen, 100kHz – 960MHz spectrum analyzer. I have been very pleasantly surprised by its performance.
It has its limitations, but it is the first usable low-cost SA I’ve found for looking at sub-GHz FM signals. There are plenty of low-cost toys out there (see my earlier post regarding the “Simple Spectrum Analyzer”) that claim to do spectrum analysis, but they have always had fatal flaws (inaccurate or horribly coarse resolution bandwidth). This one has quite good frequency and amplitude accuracy out of the box and a usable 3kHz RBW.
It has some limitations too (But for goodness sakes, we’re talking about a pocket-sized spectrum analyzer for $55!!!):
- 960MHz max frequency
- 3kHz resolution bandwidth (RBW)
- Amplitude uncertainty seems to be around 2dB
- Easily over-driven (claims +10dBm max input, but works much better below -10dBm)
Below are some pictures showing the SA in action, including both strengths and limitations.
For comparison, here is what the signal generator output looks like on a higher-end piece of test gear (Anritsu MS8609A):